Elight hair removal e-light rf hair removing skin care beauty
equipment IPL RF machine
System components description
l LCD screen: Display the work status of the system
l Emergency shut-off knob: Shut down the system on emergency
l Key switch: Turn on and shut off the power supply of the system
l Main console: house the cooling system, power supply system,
operating system and
l Treatment handle Plug: Connect the treatment handle and the main
l Treatment head supporter: Place the treatment handle when being
not in use.
l Treatment handle: Emit the intense pulse light and guide the
light to the target area.
Counter: Treatment of the working frequency of the handle
Hand Chin switch: press began firing after the IPL and RF energy
Treatment Head: IPL and RF energy output As shown below, the RF
electrode output RFenergy, Light Crystal: IPL light energy output.
Security types:Class I B The operating system uses the single-phase
230 V Power and the largest input current is not more than 6.3A.
The use of Voltage-regulator (its power is 1000W) can reduce the
influence of network fluctuation to strengthen the system‘s life.
Movement and Transportation safety
The machine contains delicate components and no allowance of long
transportation after first installation and adjustment.
1. Intense pulse light with wavelength of 560nm—1200nm, sourced
from the system, can create safety hazard. All personnel including
patients in the treatment room must wear protective glasses
whenever the unit is in use
2. To be alert that intense pulse light emission presents an eye
3. Never permit reflective objects such as jewelry, mirrors and
watches to reflect the intense pulse light.
4. Never look directly at the intense pulse light emitted from the
treatment handle. Never direct the intense pulse light at any skin
except the test patch and the target area.
5. This system has both high radio frequency and power output. Try
avoid using any other medical devices all together at the same
|light source||Intense Pulsed light|
|Transmission system||Light Guide Crystal|
|Safety Category||Class I, type B|
|Pulse Duration||See 6-2|
|The light mouth dimension||8mm×34mm,error±0.2mm|
|Surface temperature of the light mouth||≤37℃|
|Cooling system||Semiconductor refrigerator ,wind cooling, water cooling|
|Dimensions (L × W × H)||670mm×450mm×330mm|
|Fuse protector dimension||F6.3AL250V|